SATELEC has always designed tips that respect the tooth's anatomy and vibrate in perfect harmony with the handpiece.
We don't deny that a brand X tip can be cheaper to buy than a genuine SATELEC tip but we would remind you that only genuine SATELEC tips have been designed specifically for use with our generators, and we strongly recommend them. Furthermore, brand X tips can be harder than tooth enamel, which could cause undesired wear, or even etching, of the tooth's surface. That's not all - you risk damaging the screw thread and electronics of your SATELEC handpiece. Last but certainly not least, even a slight physical or electronic incompatibility could cause a brand X tip to break ... with a risk of ingestion by the patient.
Understandably, SATELEC's liability - both legal and with regard to warranty of parts and accessories - cannot be engaged for damages that arise form the use of any other than genuine SATELEC accessories.
Welcome to Acteon's product catalogue
. Non cutting instruments with parallel flutes respectful of the root canal anatomy
. Minimally invasive protocol
F43807 IRR 20-21 mm x 4
F43808 IRR 20-25 mm x 4
F43805 IRR 25-21 mm x 4
F43806 IRR 25-25 mm x 4
. Efficient cleaning and disinfection of the root canal.
. Very efficient removal of dentinal debris, pulp tissue and bacteria
. More efficient irrigant
. The shape improves micro-streaming and micro-cavitation in fluids.
. Non cutting rounded end prevents damaging the apical constriction
. Designed for the safe removal of smear layer, dentine debris and bacteria from the root canal
. Used during Passive Ultrasonic Irrigation (PUI) with Sodium Hypochlorite.
. Irrigation of the entire root canal, especially in the apical area
. PUI improves the sealing of a root canal filling
. 3 x 1 minute irrigation at the end of the root canal preparation.
. Irrisafe is inserted 1-2 mm short of the working-length.
. Irrisafe must vibrate freely in the apical area.
. Length of instruments : 21 mm or 25 mm (according model)
. Diameter : 0,20 mm or 0,25 mm (according model)
. Irrisafe instruments should always be used at a medium power setting (30-35% power).
. Color Coding System : YELLOW= Endo Mode
TIPS,ENDODONTICS IRRIGATION , SMEAR LAYER , DESINFECTION